New instrument power supply (MBTF durability rating of 200,000+ hrs), new main instrument CPU board
Rebuilt A, B, and cleave valve blocks: replaced all membranes (lifespan of 10+ years), replaced all vent valves, tested and verified all flowpaths and valves are working and there are no vacuum, air, nitrogen, or fluid leaks
Working air/vacuum pump with dual water traps for humid environments (secondary trap not shown in photos; will be mounted upon installation)
Test synthesis performed with LCMS results available upon request
Refurbished Dell Optiplex 960 (new solid state hard drive, new power supply, clean install of Windows 7 SP1, computer set up as instrument controller with Symphony 3.3.0 software); monitor, keyboard, and mouse included
On-site installation with 3 days of training and running of test synthesis included (North America only; worldwide available with extra travel cost but 220v will require a transformer for pump); installation is recommended because fittings/boards tend to loosen during shipment.